products C & tanδ (dissipation factor) measurement
Under the Tettex Instruments label we offer a broad range of precision C & tanδ measuring bridges covering both high and low voltage applications for laboratory, factory, workshop and field use.
 NEW 2830/2831 [ New Precision Oil and Solid Dielectric Analyzer ]
The Tettex 2830/2831 is the result of extensive research and years of experience testing dielectric properties of liquid and solid insulating materials. It incorporates a fast and highly advanced procedure to measure Capacitance and tan δ, DC resistance and relative permittivity e_{r} (dielectric constant) of liquid or solid insulating materials. A simple onetimeconnection system together with resistance measurement according to preselected standards drastically reduces measuring time.
The system consists of 2830, the measuring bridge and control unit, and 2831, the extension unit.
This system is a complete replacement of the Tettex 2821 and 2822.
product brochure
2840[ HighPrecision C, L & tan δ (Dissipation Factor) Measuring Bridge ]
Accuracy capacitance 0.02%, tanδ 1x10^{5} The HighPrecision DielectricLoss Analyzing System 2840 is designed for measurement of very low dielectric losses and impedances (Dissipation Factor and Power Factor) of highvoltage apparatus (e.g. extruded insulation on power cables). The instrument works on the principle of a combined bridgevectormeter and is capable of analyzing capacitive and inductive loads – especially shunt reactors – with outstanding accuracy and stability certified by a leading metrology institute. Advanced software functionalities such as insulation temperature correction, programmable test sequences with pass/fail limits, graphical visualisation of measured data, etc. make this instrument a powerful tool for analysis of highvoltage equipment.
Product Broschure
Download: ISH 2011_Paper Shunt reactor
2820a[ C, L & tan δ (Dissipation Factor) Measuring Bridge ]
Accuracy capacitance 0.1%, tanδ 1x10E4 The DielectricLoss Analyzing System 2820a is designed for measurement of low dielectric losses and impedances (Dissipation Factor and Power Factor) of highvoltage apparatus (e.g. the insulation of bushings). The instrument works on the principle of a combined bridgevectormeter and is capable of analyzing capacitive and inductive loads with high accuracy and stability. The Graphical User Interface of the instrument is highly intuitive, is focussed on convenience with builtin useful programs (e.g. support tool for tuning the external highvoltage supply) and uses a large colour touch screen as the input device.
product brochure
MIDAS micro 2883
[ Mobile insulation diagnosis & analysing system ]
The MIDAS micro 2883 is an insulation diagnosis set used to test the insulation of high voltage components i.e. transformers, bushing, capacitors etc. Designed and tested for heavy duty conditions, the complete instrument is built in one ruggedized case with casters. As it weights only 25 kg / 55 lbs it can be transported easily to the workplace  in the field or the factory. All accessories are stored in a robust duffle bag that can be attached to the case. Cable drums make setup and storing convenient and quick.  Capacitance, Dissipation / Power factor testing
 Compact onebox design with only 25 kg / 55 lb
 12 kV high voltage source
 Frequency range of 15 – 400 Hz
 Accuracy of 0.3% (capacitance) and 1x104 tan d
 Safe operation with interlock, emergency stop and safety checks
 7” colour touch screen for easy Operation
 Basic manual mode and guided Wizard mode
product brochure
MiDAS 2880, 2881
[ Mobile Insulation Diagnosis & Analysing System ]
The MIDAS 2880/2881 is an insulation diagnosis set for measuring power / dissipation factor / tan δon a variety of high voltage components in the factory and the substation. It includes a powerful and variable frequency high voltage source up to 15kV. The capabilities can be expanded with a variety of accessories to perform additional tests on transformers and rotating machines. To analyse condition and quality of high voltage insulation systems the MIDAS performs automated measurement of dissipation factor (tan δ) and power factor (cos φ), short circuit impedance and excitation current. Additional measuring capabilities like Capacitance, Quality Factor, Frequency, Voltage, Current, Power, Losses, Impedance, Inductance, Reactance, Spectrum Analyser, Digital Scope and Data Logger are included.
product brochure
Tettex procedure "Short circuit impedance with Midas"
3320[ Standard Air Capacitors ]
These standard fixed and variable capacitors are especially for precision measurements of high accuracy. Because of their high insulation resistance, small selfinductance and dissipation factor, these standards may be used within large frequency ranges up to very high frequencies.
product brochure
3370 NK[ SF6 filled high precision Standard Capacitors ]
The SF6 gas insulated standard capacitor is used together with a C and tan delta measuring bridge as the comparison standard leg of the bridge. This capacitor design is used as a calibration standard for high voltage laboratories and metrology national institutes for decades. Highest longterm stability is guaranteed and proven over 60 years.A wide range of different sizes, values and voltage ranges are available.
product brochure
3388/100/25[ Compressed gas standard Capacitor ]
The compressed gas standard capacitor is an indispensable instrument in every high voltage laboratory and test field where it occupies a wide range of important functions. The compressed gas capacitor is used as capacitance standard in bridge circuits to measure the dielectric dissipation factor tan d, the dielectric constant er respectively the capacitance of cables, ducts, insulators, transformer windings and insulates.
product brochure
3721 & 3723[ Dissipation factor tan d standards ]
Dissipation factor tan d standards are designed for checking the accuracy of capacitance and dissipation factor measuring instruments. A tan d standard is basically a standard capacitor in series with a resistor. To avoid excessive resistance values, a 1000 pF capacitor with a negligible dissipation factor should be used. The resistance units have to be calibrated together with the standard capacitor (not incl. in scope of supply).
product brochure
5286 / 5287[ Current Booster ]
Current Booster (current multiplier transformer) for MIDAS 2880 to increase the output current of the MIDAS HV supply. Used for inductance testing, especially for short circuit impedance testing of power transformers acc. IEC60076 to diagnose transformer winding deformation.
product brochure
5289 [ Resonating Inductor ]
15kV Automatic resonating Inductor used together with MIDAS for testing very high capacitance values up to 1.56uF e.g. for testing of large power generator windings. The resonator (variable inductance) operates together with the capacitive load as a parallel resonance circuit and therefore expands the possible load range of the MIDAS power source.
product brochure
5288A[ Resonating Inductor ]
10kV (*12kV) Resonating Inductor used together with MIDAS for testing very high capacitance values up to 1uF e.g. for testing of large power generator windings. The resonator (variable inductance) operates together with the capacitive load as a parallel resonance circuit and therefore expands the possible load range of the MIDAS power source.
product brochure
6835 [ Rugged Test Cell for liquid insulants ]
Rugged test cell for onsite measurements on liquid insulation samples normally used together with MIDAS 2880.The cell itself is stored in a sturdy transportation shell case.
product brochure
4761, 4764[ Standard current transformers (comparators) ]
The 4761 and 4764 high precision standard current comparators are used as comparison standard in the differential method (bridge) measurement of current instrument transformers together with the measuring instruments type2763 and 2767. The special active flux compensated design of these comparators results in negligible current errors. These units can also be used as current measuring range extension for C and tand measuring equipment.
product brochure
4762 [ Current Comparator ]
Current Comparator (current divider transformer) used together with 2820a or 2840 instrument to expand the maximal measuring current range of 15A.
product brochure
2903, 2903a [ Precision Test Cell for liquid insulants ]
This precision test cell measures the dielectric properties of liquid insulants like insulation oil etc. The test cell is designed as a vacuumsealed cylindrical capacitor (2903) with shielded measuring electrode. The cell itself is inserted into a heat able pot (2903a) with all necessary measurement connections.
product brochure
2914 [ Precision Test Cell for solid insulants ]
This precision test cell measures the dielectric properties of solid insulants like paper, plastic foil, etc. Basically the test cell consist of a heat able, shielded plate capacitor which is protected by a bell jar and can be evacuated and filled with protective gas.
product brochure
contact us
